Message 7 of This will pause the program. You should post the code you aer using now. For that we have created code. For hi-jacking thread, I was searching related to my problems, I found that one, so there only I asked my question. Message 5 of What are you actually measuring in the first frame and how is it different in the third frame?
|Date Added:||12 September 2013|
|File Size:||6.11 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Your mentioned vi is for only increment it seems, But I need within the voltage range -1v to 1v we need to measure current across the 1 kiloohm resistor, which we have connected as in Keithley Manual. So i want to finish my work with 5.
In this we are getting dynamic graph which must be linear. I don’t need your code which you didn’t attach hereI need the response from the K when requesting voltage values Its in labview 5. I m going to attach my code. Most Active Hardware Boards: How to add decimal point in out put data.
Have you got code that updates the voltage range? What are you actually measuring in the first frame and how is it different in the third frame? Message 5 of Message 8 of And about all I can tell is that keithleey sweep function is sort of working it does not start at the initial voltage and does not stop at the final voltage.
I need to work on this. Most Active Software Boards: Unless someone who has the actual instrument sees this post and offers help, you will have to dig through the instrument manual.
Message 28 of Keithley xy graph. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. If you don’t get the correct values out, then you may be programming the instrument incorrectly.
Using this we want to measure continuous readings form Message 3 of Why would you see if the function selected is greater than 4. Just try it sir, and reply me fast. Are you using the VIs which I found here, at ni.
Keithley ke Meter – IEEE (GPIB) Driver for LabVIEW – National Instruments
You don’t get it. Message 4 of Here with missing subvi,as u told its nit at all starting with initial voltage and ending at final voltage, but i m getting reading for -1 v and 1v using my labview program.
You can then manually operate the instrument and try to take a measurement. I would recommend to init the K with this settings: Sorry sir, this is only subvi.
Tektronix and Keithley
Once you determine how the instrument needs to be changed, you will have to determine how to set it up programatically. If uknow then solve it and send it to me Sir, I cant wait more for this. Message 23 of Message 24 of